title: AmScope SM-4BZ-FRL Professional Binocular Stereo Zoom Microscope, WH10x Eyepieces, 3.5X-90X Magnification, 0.7X-4.5X Zoom Objective, 8W Fluorescent Ring Light, Double-Arm Boom Stand, 110V-120V, Includes 0.5x and 2.0x Barlow Lenses
full: AmScope SM-4BZ-FRL Professional Binocular Stereo Zoom Microscope, WH10x Eyepieces, 3.5X-90X Magnification, 0.7X-4.5X Zoom Objective, 8W Fluorescent Ring Light, Double-Arm Boom Stand, 110V-120V, Includes 0.5x and 2.0x Barlow Lenses
  ['The AmScope SM-4BZ-FRL professional stereo zoom microscope has a pair of 10x super-widefield high-eyepoint eyepieces, a 0.7x-4.5x zoom objective, two Barlow lenses, and a double-arm boom stand. The microscope has an overall magnification range of 3.5x-90x. The binocular viewing head has an interpupillary range of 55 to 75mm, a 45-degree inclination to reduce eye and neck strain, and 360-degree rotation to enable sharing. The WH10x20mm super-widefield high-eyepoint eyepieces combine with the 0.7x-4.5x zoom objective to provide continuous zoom magnification and a longer working distance for inspecting large-scale specimens that require handling or repair. The microscope coms with 0.5x and 2.0x Barlow lenses that can be added to the objective to increase the magnification range. A Barlow lens with a magnification of less than 1.0 reduces magnification and increases the working distance, while a Barlow lens with a magnification greater than 1.0 increases the magnification and reduces the working distance. High-eyepoint eyepieces ease viewing for users who wear glasses, and dioptric adjustment accommodates individual eye-strength differences. A stereo microscope, sometimes called an inspection or dissection microscope, has low magnification and a long working distance that enables users to manipulate the object being inspected.', 'The removable 8W fluorescent ring light provides cool, white light for observing specimens as they would appear in their natural environment. A ring light provides shadow-free illumination. It has a 2-3/8" inner diameter and uses a 110V-120V power supply. The microscope has a double-arm boom stand that enables users to position the microscope head on all three axes (X, Y, and Z). Bilateral focus eases use for left- and right-handed users. The stand has 20" double arms, and a 30" overall length. The pillar is 17" and the focus rack has a 3" diameter. The solid cast-steel base is 2 x 8-1/4 x 10-1/4 inches (H x W x D, where H is height, the vertical distance from the lowest to highest point; W is width, the horizontal distance from left to right; D is depth, the horizontal distance from front to back).', 'Specifications', 'HeadBinocularMagnification range3.5x-90xZoom objective power0.7x-4.5xEyepieces (DIN, 30mm)WH10x20mm high-eyepointZoom range26:1Field of view2-1/2"Optical working distanceUp to 8"Microscope standDouble-arm boom standHead movementX-, Y-, and Z-axesIllumination typeEpiscopic (reflected)Light sourceFluorescent ring, 8WPower110V-120V', 'Microscopes are instruments used to enhance the resolution of an object or image. Types include compound, stereo, or digital. Compound microscopes use a compound optical system with an objective lens and an eyepiece. Stereo microscopes show object depth in a three-dimensional image. Digital microscopes are used to display an image on a monitor, rather than looking through a lens. Microscopes can have monocular (one), binocular (two), or trinocular (three) eyepieces, with varying 
  ['Professional stereo microscope with boom stand has long working distance to enable users to perform work or manipulate large items, including circuit boards and dental appliances', 'Binocular viewing head with pair of 10x super-widefield eyepieces, adjustable interpupillary distance, fixed 45-degree vertical inclination to reduce eye and neck strain, and 360-degree rotation capability to enable sharing', '0.7x-4.5x zoom objective provides continuous zoom magnification and longer focal length for inspecting large-scale specimens, a 0.5x Barlow lens extends the working distance, and a 2.0x Barlow lens extends the magnification range', 'Removable 8W fluorescent ring in a plastic housing provides bright, shadow-free illumination', 'Double-arm boom stand has 20